| Abstract|| |
Context: Some radicular changes are challenging for clinicians to diagnose, such as of root perforations, external root resorption (ERR), and vertical root fractures (VRFs). This study aims to facilitate it by comparing the diagnostic accuracy of cone-beam computed tomography (CBCT), orthopantomography, and conventional and digital periapical radiography (DPR) in the diagnosis of such problems. Is it worth doing CBCT despite the radiation dose?
Aims: To evaluate and compare the diagnostic accuracy of CBCT, panoramic radiography, and conventional and DPR in the diagnosis of root perforation (RP), ERR, and VRF.
Materials and Methods: The sample consisted of 40 extracted human teeth and 10 macerated human mandibles. RPs were performed using diamond burs, ERRs using spherical carbide burs, and RFs using a universal machine EMIC-DL 1000. The images were evaluated by 6 dentomaxillofacial radiologists.
Results: Receiver operating characteristic (ROC) revealed that CBCT showed the highest area under the ROC curve (Az) values for RP, ERR, and VRF (0.903, 0.950, and 0.849, respectively). The worst Az values for RP, ERR, and VRF (0.718, 0.494, and 0.611, respectively) were for panoramic radiography.
Conclusions: CBCT showed the best results in the diagnosis of ERR and VRF. The diagnosis of ERR was the least accurate, panoramic radiography being not appropriate for its diagnosis. CBCT and conventional periapical radiography obtained similar results for the evaluation of RP. So for, RP indicate the conventional periapical radiography because CBCT has a higher radiation dose.
Keywords: Cone-beam computed tomography, diagnosis, digital radiography, orthopantomography, root resorption, tooth fractures
|How to cite this article:|
Takeshita WM, Chicarelli M, Iwaki LC. Comparison of diagnostic accuracy of root perforation, external resorption and fractures using cone-beam computed tomography, panoramic radiography and conventional & digital periapical radiography. Indian J Dent Res 2015;26:619-26
The diagnosis of radicular changes is a challenge in dentistry, one that requires imaging techniques that allow observation not only of the tooth in question but also of the adjacent bone structures. ,
|How to cite this URL:|
Takeshita WM, Chicarelli M, Iwaki LC. Comparison of diagnostic accuracy of root perforation, external resorption and fractures using cone-beam computed tomography, panoramic radiography and conventional & digital periapical radiography. Indian J Dent Res [serial online] 2015 [cited 2019 Oct 16];26:619-26. Available from: http://www.ijdr.in/text.asp?2015/26/6/619/176927
Dental radiology offers several imaging methods to examine anatomical and pathological details during treatment planning. However, despite the advancements in the area, the diagnosis of some radicular changes, such as root perforation, external root resorption (ERR) and vertical root fractures (VRFs), can still be a difficult task for dentists. 
Root perforation (RP) is a communication between the root canal system and the external root surface.  Usually an iatrogenic and undesirable incident, RP may occur at any stage of endodontic treatment and represent 10% of endodontic failures. ,
ERR has a complex etiology and its classification depends on the location and stimuli that induce its development. ERR in the permanent dentition usually results of trauma, orthodontic movements, tumors, and eruption of teeth.  Early imaging diagnosis is essential since the stimulus that causes resorption must be removed as soon as changes are observed.
VRF are caused by impact of great strength. Compression on lingual and/or buccal aspect leads to separation of the root into two or more fragments. , The diagnosis of VRF is difficult and imposes a challenge for the treatment planning.  Clinical signs and symptoms of VRF are not pathognomonic of the condition and its prognosis is poor. ,,
Conventional and digital periapical radiography (DPR) and panoramic radiography are typically used to diagnose radicular changes. Although they offer acceptable images in the mesiodistal aspects, they are unsatisfactory in the buccolingual aspect;  a problem minimized with the advent of three-dimensional (3D) images of cone-beam computed tomography (CBCT). ,,,,
Thus, the aim of the present study was to compare CBCT diagnostic accuracy with that of panoramic radiography and conventional and DPR in the diagnosis of RP, ERR, and VRF in the middle third of the root.
| Materials and Methods|| |
The study was approved by the Standing Committee on Ethics in Human Research of the State University of Maringá-PR, Brazil (Protocol CAAE: 03731012.9.0000.0104). This research was conducted in accordance with the Declaration of Helsinki.
The sample consisted of 40 premolars teeth extracted from the Human Teeth Bank of the Department of Dentistry, State University of Maringá. The teeth were mounted in alveolar sockets of 10 macerated human mandibles using pink wax 7 (Polidental, Cotia, São Paulo, Brazil); hence, they were kept in place but could also be removed to create the artificial defects.
Artificial ERR were performed by the same operator using spherical carbide burs (¼ diameter, KG Sorensen, Cotia, SP, Brazil), on the middle third of the three in the distal, three in the mesial, two in the lingual, and two in the buccal aspects of the roots, perpendicularly positioned to the surface, in high-speed rotation and constant cooling. , The size of the wear was 0.25 mm diameter/0.25 mm depth holes correspond to the size of the carbide burs. Only one size was used. The small size of resorption was determined based on literature, including being smaller than reported in literature. , For minor defects, the greater the difficulty of diagnosis. ,
RPs were produced by single operator using diamond drills (FG 1011HL, kG Sorensen, Cotia, SP, Brazil), in a water-cooled high-speed handpiece at 45° angle to the long axis of each tooth, until perforations were observed on the two in the lingual, two in the buccal, three in the mesial, and three in the distal aspects of the root.  The size of the perforations was 0.5 mm diameter. Only one size was used. The size was based on the literature and is equivalent to the size of the drill used in coronary opening, a major cause of RPs. 
For the creation of VRFs, first, all teeth were examined using a stereomicroscope (Olympus SZX7, São Paulo, SP, Brazil). The crowns were sectioned 2 mm above the dentin-enamel junction. ,, The teeth were placed into perforated polyvinyl chloride pipes to give the assembling material some retention and were fixed in place with acrylic resin (Classic, São Paulo, SP, Brazil).
A universal testing machine (EMIC-DL 1000, EMIC, São José dos Pinhais, PR, Brazil) created the VRFs using adequate strength for static tensile and compression tests. The strength was applied vertically to the center of the teeth, which were slightly worn to adapt the machine's strength applicator tip. Root fractures were generated in the vertical plane and the fracture line inclined orientation (buccolingual or mesiodistal) was also recorded. According to a protocol reported in literature, ,, tooth roots broke into more than three fragments and were excluded from the study.
The same control group was used for comparison of RP-group and ERR-group. To compare with the VRF-group, the control group used was previously sectioned crown 2 mm above the dentin-enamel junction. ,, The teeth were divided into four groups with 10 teeth each: Control Group, RP-Group, ERR-Group, and VRF-Group. Sample size determination specified the sample size used in the present study  as well as a pilot study [Table 1].
|Table 1: Sample size determination based on literature and a pilot study |
Click here to view
The teeth mounted in the jaws were radiographed with conventional periapical radiographs (CPR) using RINN film holder (Dentsply ® , York, PA, USA). Three radiographs were taken with different beam angulations: An orthogonal projection and two projections with 20° horizontal shifts to mesial and distal of the central X-ray. ,
DPR were taken using RINN film holder (Dentsply ® , York, PA, USA) coupled to complementary metal-oxide semiconductor sensor (CMOS, Schick ® , Long Island City, New York, USA), again with different beam angulations: An orthogonal projection and two projections with 20° horizontal shifts to mesial and distal of the central X-ray. ,,
Radiographs were taken using a dental X-ray machine 70X (Dabi Atlante ® , São Paulo, Brazil) at 70 kVp, 8 mA, cylindrical locator with 40 cm focal length, exposure time of 0.4 s. Exposure parameters had been previously piloted according to the as low as reasonably achievable principle (as low as reasonably achievable).
Panoramic radiographs were obtained with Orthoralix 9200 GENDEX (Dentsply ® , York, PA, USA) using T-Mat G/RA Kodak film (15 cm × 30 cm, Carestream Health ® , Rochester, New York, USA) with their metallic cassette and intensifying screen (Kodak Lanex Medium Extraoral Imaging Screens X-Omat). The midsagittal plane of the jaws was positioned perpendicularly to the ground and the occlusal plane parallel to the ground. Periapical and panoramic films were processed in an automatic processor (Revell X-TEC, São Paulo, Brazil).
CBCT images were obtained with an i-CAT scanner (Imaging Sciences International, Hatfield, PA, USA), using 14-bit grayscale, field of view of 6 cm, voxel of 0.125 mm, 120 kVp, and 36.2 mAs exposure time, with the midsagittal plane of the jaws positioned perpendicularly to the ground and the occlusal plane parallel to the ground.
The CBCT images and DPR images were observed in an Intel Core I7 computer (6-GHz RAM, 500 GB hard drive) with a 20 inch LCD monitor (AOC, Top Victory Electronics, Taiwan), with the software CDR Dicom for Windows V4.5 (Schick Technologies, Long Island City, New York, USA) and the software Xoran CAT (Xoran CAT V.2.0.21, Xoran Technologies, Ann Arbor, MI, USA). The observers, according to necessity, adjusted the images' brightness and contrast, utilized the zoom tool, same room and evaluated the images from all available plans and reconstructions (axial, sagittal, and coronal planes; panoramic and parasagittal reconstructions). Conventional radiographs were examined on a light viewing box, in a dark room, using a × 4 magnifying glass. All the images were evaluated by six experts in Dental Radiology, and root changes were classified using a five-point scale: (1) Change definitely present, (2) change probably present, (3) uncertain, (4) change probably absent, and (5) change definitely absent. Random images were evaluated the 2 nd time 15 days later (Takeshita et al.;  Takeshita et al.  ).
Statistical analysis used receiver operator characteristic (ROC) curves to evaluate the accuracy of the imaging methods. The values obtained for areas under the ROC curve were compared in pairs, with a significance level of 0.05 (MedCalc 12.3.0 Software, Ostend, Belgium). Kappa test  was used to verify intra-examiner reliability, which revealed very good intra-examiner agreement, except for good intra-examiner agreement for panoramic radiography [Table 2].
| Results|| |
CBCT obtained the highest values of sensitivity and specificity of the area under the ROC curve (Az) whereas the lowest were for panoramic radiography [Table 3] and [Figure 1]. The mean Az values of CBCT differed significantly from DPR and from panoramic radiography [Table 4].
|Figure 1: The receiver operator characteristic curve for the different imaging methods that analyzed root perforation|
Click here to view
|Table 3: Area under the receiver operating characteristic curve and standard deviation of root perforation for each imaging method |
Click here to view
|Table 4: Comparison of the area under the receiver operating characteristic curve among the imaging methods for root perforation |
Click here to view
External root resorption
Again, sensitivity and specificity of the area under the ROC curve (Az) were high for CBCT and low for panoramic radiography [Table 5] and [Figure 2]. The mean Az values of CBCT differed significantly from CPR, DPR, and panoramic radiography [Table 6].
|Figure 2: The receiver operator characteristic curve for the different imaging methods that analyzed external root resorption|
Click here to view
|Table 5: Area under the receiver operating characteristic curve and standard deviation of external root resorption for each imaging method |
Click here to view
|Table 6: Comparison of the area under the receiver operating characteristic curve among the imaging methods for external root resorption |
Click here to view
Vertical root fracture
Once again, CBCT showed the highest values of sensitivity and specificity of the area under the ROC curve (Az), and panoramic radiography the lowest ones [Table 7] and [Figure 3]. The mean Az values of CBCT differed significantly from CPR, DPR, and panoramic radiography [Table 8]. In addition, CPR differed from DPR.
|Figure 3: The receiver operator characteristic curve for the different imaging methods that analyzed vertical root fracture|
Click here to view
|Table 7: Area under the receiver operating characteristic curve and standard deviation of vertical root fracture for each imaging method |
Click here to view
|Table 8: Comparison of the area under the receiver operating characteristic curve among the imaging methods for vertical root fracture |
Click here to view
| Discussion|| |
A careful treatment planning uses several imaging methods to evaluate anatomical and pathological details the oral and maxillofacial complex. However, despite the current improvement in imaging techniques, diagnosis of some radicular changes, such as RP, ERR, and VRF, is still challenging. 
Several instruments and techniques are suggested for clinical diagnosis of RP such as electronic apex locator, surgical microscopy, endoscopy, and radiographic methods. ,, The present study evaluated RP using imaging methods recommended in literature: CPR, DPR, panoramic radiography, and CBCT. The RP was performed in different positions with the purpose of simulating the different possible failures in endodontic treatment.  Similarly to the previous research, ,, the present study found that CBCT was the most accurate imaging method.
CBCT was the most accurate in the diagnosis of RP in the present study, showing a higher value than that found by Shemesh et al.  The present study used teeth without endodontic treatment as it is believed that once RP is found, imaging examinations should be performed to evaluate the alteration and to avoid endodontic treatment-a questionable treatment for this problem. In contrast, the result of present result is similar to that found by D'Addazio et al.  probably because the authors also examined RP using teeth without endodontic treatment [Figure 4].
|Figure 4: Root perforation images. (a) Cone-beam computed tomography. (b) Digital periapical radiography using parallax principle. (c) Conventional periapical radiograph using parallax principle. (d) Panoramic radiography|
Click here to view
The etiology of ERR is complex and its classification depends on the location and stimulus that can cause its development. Hence, the EER was held in different positions with the purpose of proving the difficulty of EER view, especially in two-dimension (2D) techniques. However, does this difficulty compensate the radiation dose received in 3D exams? Orthodontic treatment is one of the etiologies of ERR, which causes rounding of the root apex. , This type of resorption occurs between 39% and 99% of patients under orthodontic treatment , as a result of a complex combination of individual biological characteristics and mechanical forces employed during treatment.
The most accurate imaging method in the diagnosis of ERR was again CBCT, with an Az value similar to those obtained by other studies, ,, and statistically different from all the other imaging methods examined in the present study. Panoramic radiography had the lowest Az value, being thus the worst method to investigate resorption the middle third of the root, whereas CBCT is the method of choice to detect ERR, particularly on the buccal and lingual aspects of the root. Although there was no statistical difference between CPR and DPR in the diagnosis of ERR, it seems that DPR leans toward a more accurate result.
Among all radicular alterations, ERR was the most difficult to detect with 2D imaging methods, as shown by the low Az values of DPR, CPR, and panoramic radiography, given that alterations in the buccal and lingual aspects of the root are more difficult to visualize with such methods. 
A single defect size was produced because the aim was to evaluate the different imaging methods [Figure 5].
|Figure 5: External root resorption images. (a) Cone-beam computed tomography. (b) Digital periapical radiography using parallax principle. (c) Conventional periapical radiograph using parallax principle. (d) Panoramic radiography|
Click here to view
Accurate diagnosis of VRF depends on a careful clinical examination, meticulous evaluation of the case, and imaging examination that allows the assessment of both bone and tooth structures. On one hand, periapical radiographs may not be useful in the diagnosis of VRF, particularly when the X-ray beam is perpendicular to the line of fracture.  On the other hand, CBCT allows the assessment not only of the fracture but also of the bone loss due to inflammation. 
Some studies indicate that CBCT is the most accurate imaging method to detect VRF for teeth either with endodontic treatment , or without it.  Case reports with two  and fours cases  demonstrated that CBCT was more accurate than CPR to diagnose VRF, showing the fracture lines clearly, and a study with a series of clinical cases  demonstrated that CBCT provided additional information about the lesion, which can aid diagnosis of VRF and help to prevent unnecessary treatment.
Some studies, however, show conflicting results regarding the accuracy of CBCT. On one hand, Edlund et al.,  who examined 32 teeth suspected of VRF from 29 patients, found that CBCT shows high accuracy and specificity in the diagnosis of VRF. Bernardes et al.  examined suspicion of VRT in 20 patients and found that CBCT was an excellent diagnostic method, superior to CPR. An in vitro study Mora et al.  also revealed that CBCT was more accurate to detect VRT than CPR. Comparing multidetector computed tomography (MCT), CBCT, and DPR to detect VRF, Khedmat et al.  found that MCT was better than CBCT as the presence of Gutta-percha reduced accuracy, sensitivity, and specificity of the latter but not of the former. However, the use of MCT is restricted due to its higher radiation dose compared to CBCT.
The results of the present study corroborate those of Bernardes et al.,  Bernardes et al.,  Bornstein et al.,  Edlund et al.,  Tang et al.,  and Takeshita et al.  given that CBCT was more accurate to detect VRF than CPR. Conversely, the results contrast with those of Kambungton et al.  and da Silveira et al.  maybe because da Silveira et al.  examined endodontically treated teeth and Kambungton et al.  used X-ray equipment very different from those used in the present study. Furthermore, in the present study, CBCT showed significant differences compared to the other imaging methods while DPR was statistically different from CPR, CBCT being more efficient in the detection of VRF [Figure 6].
|Figure 6: Vertical root fracture images. (a) Cone-beam computed tomography. (b) Digital periapical radiography using parallax principle. (c) Conventional periapical radiograph using parallax principle. (d) Panoramic radiography|
Click here to view
Although different imaging methods are important in the diagnosis of VRF, a systematic review  on the matter asserts that the studies carried out so far lack precision and that the existing data, with their clinical and radiographic indices, are still insufficient to detect VRF accurately. The authors  argue that the despite the advent of CBCT, the diagnosis of VRF is still challenging, even for experienced professionals.
| Conclusion|| |
The present study found that:
- CBCT images were more accurate than DPR, CPR, and panoramic radiography to detect RP, ERR, and VRF
- Panoramic radiography was the least accurate imaging method to detect radicular changes
- ERR was to most difficult radicular change to diagnose
- Panoramic radiography was not a suitable method to detect ERR in the middle third of the root
- CBCT and conventional periapical radiography obtained similar results for the evaluation of RP.
Financial support and sponsorship
Conflicts of interest
There are no conflicts of interest.
| References|| |
Pretty IA, Maupomé G. A closer look at diagnosis in clinical dental practice: Part 3. Effectiveness of radiographic diagnostic procedures. J Can Dent Assoc 2004;70:388-94.
Bernardes RA, de Moraes IG, Húngaro Duarte MA, Azevedo BC, de Azevedo JR, Bramante CM. Use of cone-beam volumetric tomography in the diagnosis of root fractures. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2009;108:270-7.
Tamse A. Vertical root fractures in endodontically treated teeth: Diagnostic signs and clinical management. Endod Top 2006;13:84-94.
American Association of Endodontists. Glossary of Endodontic Terms. 7 th
ed. Chicago: American Association of Endodontists; 2003.
de Chevigny C, Dao TT, Basrani BR, Marquis V, Farzaneh M, Abitbol S, et al.
Treatment outcome in endodontics: The Toronto study - Phase 4: Initial treatment. J Endod 2008;34:258-63.
Shemesh H, Cristescu RC, Wesselink PR, Wu MK. The use of cone-beam computed tomography and digital periapical radiographs to diagnose root perforations. J Endod 2011;37:513-6.
Bernardes RA, de Paulo RS, Pereira LO, Duarte MA, Ordinola-Zapata R, de Azevedo JR. Comparative study of cone beam computed tomography and intraoral periapical radiographs in diagnosis of lingual-simulated external root resorptions. Dent Traumatol 2012;28:268-72.
Welbury R, Kinirons MJ, Day P, Humphreys K, Gregg TA. Outcomes for root-fractured permanent incisors: A retrospective study. Pediatr Dent 2002;24:98-102.
Bornstein MM, Wölner-Hanssen AB, Sendi P, von Arx T. Comparison of intraoral radiography and limited cone beam computed tomography for the assessment of root-fractured permanent teeth. Dent Traumatol 2009;25:571-7.
Edlund M, Nair MK, Nair UP. Detection of vertical root fractures by using cone-beam computed tomography: A clinical study. J Endod 2011;37:768-72.
Tamse A, Kaffe I, Lustig J, Ganor Y, Fuss Z. Radiographic features of vertically fractured endodontically treated mesial roots of mandibular molars. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2006;101:797-802.
Krell KV, Rivera EM. A six year evaluation of cracked teeth diagnosed with reversible pulpitis: Treatment and prognosis. J Endod 2007;33:1405-7.
Opdam NJ, Roeters JJ, Loomans BA, Bronkhorst EM. Seven-year clinical evaluation of painful cracked teeth restored with a direct composite restoration. J Endod 2008;34:808-11.
Takeshita WM, Vessoni Iwaki LC, Da Silva MC, Filho LI, Queiroz Ade F, Geron LB. Comparison of the diagnostic accuracy of direct digital radiography system, filtered images, and subtraction radiography. Contemp Clin Dent 2013;4:338-42.
Tsurumachi T, Honda K. A new cone beam computerized tomography system for use in endodontic surgery. Int Endod J 2007;40:224-32.
Kobayashi K, Shimoda S, Nakagawa Y, Yamamoto A. Accuracy in measurement of distance using limited cone-beam computerized tomography. Int J Oral Maxillofac Implants 2004;19:228-31.
Ludlow JB, Davies-Ludlow LE, Brooks SL, Howerton WB. Dosimetry of 3 CBCT devices for oral and maxillofacial radiology: CB Mercuray, NewTom 3G and i-CAT. Dentomaxillofac Radiol 2006;35:219-26.
Bissoli CF, Ágreda CG, Takeshita WM, Castilho JC, Medici Filho E, Moraes ME. Importancia y aplicaciones del sistema de tomografia computarizada cone-beam (TCCB). Acta Odontol Venez 2007;45:589-92.
Roberts JA, Drage NA, Davies J, Thomas DW. Effective dose from cone beam CT examinations in dentistry. Br J Radiol 2009;82:35-40.
Westphalen VP, Gomes de Moraes I, Westphalen FH, Martins WD, Souza PH. Conventional and digital radiographic methods in the detection of simulated external root resorptions: A comparative study. Dentomaxillofac Radiol 2004;33:233-5.
Melo SL, Bortoluzzi EA, Abreu M Jr., Corrêa LR, Corrêa M. Diagnostic ability of a cone-beam computed tomography scan to assess longitudinal root fractures in prosthetically treated teeth. J Endod 2010;36:1879-82.
Ferreira RI, Bahrami G, Isidor F, Wenzel A, Haiter-Neto F, Groppo FC. Detection of vertical root fractures by cone-beam computerized tomography in endodontically treated teeth with fiber-resin and titanium posts: An in vitro
study. Oral Surg Oral Med Oral Pathol Oral Radiol 2013;115:e49-57.
Takeshita WM, Iwaki LC, da Silva MC, Sabio S, Albino PR. Comparison of periapical radiography with cone beam computed tomography in the diagnosis of vertical root fractures in teeth with metallic post. J Conserv Dent 2014;17:225-9.
Lemeshow S, Hosmer D. Adequacy of Sample Size in Health Studies. Chichester: John Wiley; 1990.
Kambungton J, Janhom A, Prapayasatok S, Pongsiriwet S. Assessment of vertical root fractures using three imaging modalities: Cone beam CT, intraoral digital radiography and film. Dentomaxillofac Radiol 2012;41:91-5.
Landis JR, Koch GG. The measurement of observer agreement for categorical data. Biometrics 1977;33:159-74.
Gordon MP, Chandler NP. Electronic apex locators. Int Endod J 2004;37:425-37.
Kim S, Kratchman S. Modern endodontic surgery concepts and practice: A review. J Endod 2006;32:601-23.
Moshonov J, Michaeli E, Nahlieli O. Endoscopic root canal treatment. Quintessence Int 2009;40:739-44.
D′Addazio PS, Campos CN, Özcan M, Teixeira HG, Passoni RM, Carvalho AC. A comparative study between cone-beam computed tomography and periapical radiographs in the diagnosis of simulated endodontic complications. Int Endod J 2011;44:218-24.
Eskandarloo A, Mirshekari A, Poorolajal J, Mohammadi Z, Shokri A. Comparison of cone-beam computed tomography with intraoral photostimulable phosphor imaging plate for diagnosis of endodontic complications: A simulation study. Oral Surg Oral Med Oral Pathol Oral Radiol 2012;114:e54-61.
McNab S, Battistutta D, Taverne A, Symons AL. External apical root resorption following orthodontic treatment. Angle Orthod 2000;70:227-32.
da Silveira HL, Silveira HE, Liedke GS, Lermen CA, Dos Santos RB, de Figueiredo JA. Diagnostic ability of computed tomography to evaluate external root resorption in vitro
. Dentomaxillofac Radiol 2007;36:393-6.
Brezniak N, Wasserstein A. Root resorption after orthodontic treatment: Part 1. Literature review. Am J Orthod Dentofacial Orthop 1993;103:62-6.
Reukers E, Sanderink G, Kuijpers-Jagtman AM, van′t Hof M. Assessment of apical root resorption using digital reconstruction. Dentomaxillofac Radiol 1998;27:25-9.
Durack C, Patel S, Davies J, Wilson R, Mannocci F. Diagnostic accuracy of small volume cone beam computed tomography and intraoral periapical radiography for the detection of simulated external inflammatory root resorption. Int Endod J 2011;44:136-47.
Zou X, Liu D, Yue L, Wu M. The ability of cone-beam computerized tomography to detect vertical root fractures in endodontically treated and nonendodontically treated teeth: A report of 3 cases. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2011;111:797-801.
Tang L, Zhou XD, Wang Y, Zhang L, Zheng QH, Huang DM. Detection of vertical root fracture using cone beam computed tomography: Report of two cases. Dent Traumatol 2011;27:484-8.
da Silveira PF, Vizzotto MB, Liedke GS, da Silveira HL, Montagner F, da Silveira HE. Detection of vertical root fractures by conventional radiographic examination and cone beam computed tomography - An in vitro
analysis. Dent Traumatol 2013;29:41-6.
Wang P, Yan XB, Lui DG, Zhang WL, Zhang Y, Ma XC. Detection of dental root fractures by using cone-beam computed tomography. Dentomaxillofac Radiol 2011;40:290-8.
Wang P, He W, Sun H, Lu Q, Ni L. Detection of vertical root fractures in non-endodontically treated molars using cone-beam computed tomography: A report of four representative cases. Dent Traumatol 2012;28:329-33.
Fayad MI, Ashkenaz PJ, Johnson BR. Different representations of vertical root fractures detected by cone-beam volumetric tomography: A case series report. J Endod 2012;38:1435-42.
Mora MA, Mol A, Tyndall DA, Rivera EM. In vitro
assessment of local computed tomography for the detection of longitudinal tooth fractures. Oral Surg Oral Med Oral Pathol Oral Radiol Endod 2007;103:825-9.
Khedmat S, Rouhi N, Drage N, Shokouhinejad N, Nekoofar MH. Evaluation of three imaging techniques for the detection of vertical root fractures in the absence and presence of gutta-percha root fillings. Int Endod J 2012;45:1004-9.
Tsesis I, Rosen E, Tamse A, Taschieri S, Kfir A. Diagnosis of vertical root fractures in endodontically treated teeth based on clinical and radiographic indices: A systematic review. J Endod 2010;36:1455-8.
Prof. Wilton Mitsunari Takeshita
Department of Dentistry, Federal University of Sergipe, Sāo Cristóvāo
Source of Support: None, Conflict of Interest: None
[Figure 1], [Figure 2], [Figure 3], [Figure 4], [Figure 5], [Figure 6]
[Table 1], [Table 2], [Table 3], [Table 4], [Table 5], [Table 6], [Table 7], [Table 8]